When i attached the probes together,the display shows "",and when i separate them,instead of showing infinity,the display is shown increasing value of resistance again,the probes do not touch anything.
It doesn't response when i try to measure a resistor and if it does the value are far wrong.
It happens after i did a capacitor test,while connecting the dmm probes ohm meter function to the capacitor to charge it a bit.
Maybe the capacitor damage something in the dmm,but when i opened the DMM,i didn't notice any burned or damage component. The battery is good,btw. What can be the component that caused this failure? Thanks in advance. Scroll to continue with content. Check the internal fuse. You can get proper meters for testing capacitors - but even with those a charged capacitor can damage the meter. A DMM doesn't seem to me like it gives any sensible reading on a capacitor, with an analogue meter the needle gives a kick as the test voltage charges the capacitor.
An analogue meter is a little more tolerant of unintentionally being used to discharge a charged capacitor, but you can still bend the needle if it hits the end stop hard enough!
The internal fuse in that DMM is not related to the resistance function,but to the ammeter function low DC currents. This DMM can not read capacitor value,but it can give you some idea on the condition of the capacitor.
But,my question are:What can be the component that caused this failure? Can you take really good clean crisp pictures of the internal components of this meter both front and back? The more the better and post them back here. I attached some photo of the internal componenets. I tried several times to take a good clean crisp picture,but the best that i got are the ones that i attached. And they,unfortunately,still bad refer to the fact that it makes it difficult to follow the circuit and the components condition.
But this is the maximum that i could achieved. Maybe the photos will be useful somehow,even though,they are not clear enough. Nevertheless,i assumed that my description about the failure will be enough,in order to get an answer.
Only the ohm meter function is failed,so only small part of the DMM should be suspected the component that related to the ohm meter. And i assumed that the scenario that i described before,will be enough to an experienced professional electrician to know what caused this scenario. Dodgydave Joined Jun 22, 8, Dodgydave said:. ISB Joined May 21, 1,Page of 13 Go. Quick Links. Table of Contents. DT Series. Page 2: Warranty If the defect has been caused by a misuse or abnormal operation conditions, the repair will be billed at a nominal cost.
In the latter case, the transient stresses are variable; for that reason, its necessary that the transient-withstand -capability of equipment is made known to the user. Page 4: Maintenance CAUTION To avoid possible damage to the meter or to the equipment under test, follow these guidelines: Disconnect circuit power and discharge all high-voltage capacitors before testing resistance, continuity, diodes, or capacitance. Use the proper terminals, function, and range for your measurements.
Before measuring current, check the meter's fuses and turn power off to the circuit before connecting the meter to the circuit. Page 5: Front Panel Description This function allows the meter to output a signal as a signal generator. Page 6 To use safely, each measurement can not last for more than 10 seconds, and the interval between each measurement must be more than 15 minutes.
Page 8 3.
Set the range switch to the desired range. If the resistor to be measured is connected to a circuit, disconnect circuit power and discharge all capacitors before measuring resistance. Page 11 1. Set the range switch to " " or " " range.
A test signal will appear between the and "COM" jacks. The voltage is approximate 5Vp-p. A coupling capacitor should be used when connecting the meter to a circuit. Page Battery And Fuse Replacement Before attempting to open the case, always be sure that test leads have been disconnected from measurement circuits.
Close case and tighten screws completely before using the meter to avoid electrical shock hazard. This manual is also suitable for: a b d e Show all h bl.
Comments to this Manuals Your Name:. Enter text from picture:. Latest comments:. Print page 1 Print document 13 pages. Cancel Delete. Sign in OR.Forums New posts Search forums.
DT830B Digital Multimeter
Articles Top Articles Search resources. Members Current visitors. Log in Register. Search titles only. Search Advanced search…. New posts. Search forums. Log in. Welcome to our site!
haoyue dt830d multimeter circuit diagram
It's a simple multimeter capable of accomplishing varied measurements. In all likelihood it contains a custom chip that performs most of the functions and it's very unlikely you'll find a circuit diagram.See more product details.
Small multimeter for your day to day needs. If you have basic knowledge about multimeters you can try this small multimeter for basic continuity checks and voltage measurements.
Resistor ohms. Transistor HFE. Square wave output. Battery test. Display: 4. Would you like to tell us about a lower price? To avoid possible damage to the meter or to the equipment under test, follow these guidelines: Disconnect circuit power and discharge all high-voltage capacitors before testing resistance, continuity, diodes, or capacitance.
Use the proper terminals, function, and range for your measurements. Before measuring current, check the meter's fuses and turn power off to the circuit before connecting the meter to the circuit.
Before rotating the range switch to change functions, disconnect test leads from the circuit under test. Before attempting to insert transistors for testing, always be sure that the test leads have been disconnected from any measurement circuits. Remove test leads from the meter before opening the Meter case. Skip to main content. Currently unavailable.
Thread Starter wdevasurendratp Joined Dec 11, 3. Scroll to continue with content. Dodgydave Joined Jun 22, 8, Does this site help in any way??
Dodgydave said:. I have one of these at work. If you can wait until tomorrow I can open it up and have a look inside. I will get back to you. Superklin Joined Jul 9, 5. Hi, I have the same problem, someone know the value of R5? MrChips Joined Oct 2, 20, Yasu Joined Mar 11, 1. What is the value of R6 resistor you had changed pls resond.
DarthVolta Joined Jan 27, The chip in all these is something like ICL, so most of the circuits are pretty similar, depending on what the cheap chipmakers cooked up that week.
Then I got another and fried it when I was careless. I only wanted that 1 as a basic ampmeter, so I might get another 1 yet. You must log in or register to reply here.HOW TO REPAIR DIGITAL MULTIMETER !!
You May Also Like. Continue to site. Yesterday at PM. Integrating a digital counter in a circuit. Apr 7, General Electronics Chat. Apr 5, Mar 28, Mar 24, Products Buying Leads Companies. Circuit Diagram of motor teaching model Jul 30, Circuit Diagram of motor teaching model. Printed circuit board assembly Apr 01, Manufacturing capability-PCB Sizexcm max Supplier: Dongguan Onlitex Electronic Co.
Supplier: Guangzhou Waytronic Electronics Co. Only one OEM board design needed. Mercedes-Benz C4 star scanner Apr 07, Complete maintenance data. It could offer full disassembly and assembly circuit diagramworking principle and element place anytime needed. It is the Supplier: Fenghua Chenpu Electronic Co. Par 56 Circuit Diagram Translation about the words in the diagram : this circuit should be used under the condition of The remote controller can control eleven kinds of effects.
Button and function: Translation Supplier: Shenzhen Ledlighting Technology Co. In this situation, we will offer circuit diagramand we will guide them how to repair local. Supplier: Supremevalue International Co. Automobile Starting System Trainer Jan 25, A color circuit diagram on the instruction board is protected by 3 mm You can select three fault types including "Open circuit fault", "Short circuit fault", and "Improper contact".
Thread starter carolb4 Start date Feb 20, Search Forums New Posts. Thread Starter carolb4 Joined Feb 20, 1. I have a Commercial Electric handheld digital multimeter. I'm trying to test parts of an icemaker and was told from this website to: Conduct this test with the refrigerator unplugged. Test the mold heater for resistance using a multitester. Set the multitester to the ohms setting X1.
Place one probe into the hole labeled "H" and the other probe into the hole labeled "L". The multitester should read in the range of 60 to 90 ohms. This multimeter does not have X1 on the meter scale. There is2K, 20K, K, etc. I don't know how to test. Please advise. Scroll to continue with content.
The scale reads to ohms. If is the lowest scale then that is the one to use. X1 refers to the "times 1" scale. I have replaced two icemakers over the years. It is my humble opinion that you will be better served to replace the icemaker with a new one. You will benefit in the long run. The nomenclature "X1" is obsolete and comes from the days when multimeters were analog devices. The basic idea was that you multiplied the resistance reading on the analog meter scale by the factor after the X.
Modern digital meters require no such conversions on the part of the user other than, perhaps, knowing about SI prefixes. VoodooMojo Joined Nov 28, SI index reference. SI Prefix. Don't you hate jokes you gotta explain. Wendy Joined Mar 24, 21, Attoboy is faint praise indeed.